Probelogic Exclusive : We repair medical ultrasound transducer. Contact us for probe care, repair, testing, replacements.
It is a Siemens/Acuson V5Ms TOE/TEE probe in which the customer was complaining about poor image quality. Our acoustic report suggests the probe has approximately 10% of elements non firing as can be seen in the image below.
The issue is almost certainly caused by a significant knock to the head of the probe as can be seen in the image below.
Now if we compare a good V5Ms ultrasound image from the same phantom on the same ultrasound system. Firstly the good image, you can see how clear the phantom dots throughout the image are:
With such gradual deterioration, it is sometimes difficult to recognize the problem as it just gets more and more difficult to scan patients. Get your probes acoustically tested today to be sure.
Learn more on Acuson/Siemens V5Ms Cardiac Phased array probe image degradation
Contact Probelogic today
Opening Hours : Monday to Friday - 8:30am to 5:00pm
Contact : +1300-611-503